Tests For: Nickel in Natural and Treated Water
Tests Range: 0–10 mg/L
Reagent Chemistry Used: Nioxime
Method Detection Limit*: 0.04 mg/L
Limit of Quantification**: 0.14 mg/L
Nickel does not occur naturally in water but is found in many industrial waste waters, such as those from the steel and plating industries. It is considered an undesirable constituent of water, and hence requires close and careful monitoring. The EU maximum admissible concentration for drinking water (MAC) is 0.02 mg/L (EFSA 2015).
The Palintest Nickel test method provides a simple test for the determination of nickel in water over the range 0–10 mg/L Ni. The test responds to both Ni2+ and Ni4+ and thus gives a measure of total soluble inorganic nickel content of the water.
In the Palintest Nickeltest method, nickel salts are reduced to the nickelous form and then reacted with nioxime indicator to give a pink coloured complex.
The reagents are provided in tablet form and the test is simply carried out by adding tablets to a sample of the water. The intensity of colour produced in the test is proportional to the nickel concentration and is measured using a Palintest Photometer.
Reagents are included to prevent copper interference, and a complexing powder is provided to prevent iron interference.
The presence of cobalt at 0.5 mg/L gives a positive response in the test.
The presence of significant levels of EDTA (at least 25 mg/L) complexes nickel and reduces response in the test. Complexing agents used in water treatment,